I'm currently working on a little DAQ system with four DRS evaluation boards. Do i need to apply any specific settings when using the clock in/out connectors for synchronization? I do not see anything like that in the drs_exam_multi example.
Any help would be greatly appreciated!
Right, I did not yet put any code there. What you need on all slave boards is
Set SetFrequency() is needed to restart the boards with the external clock.
This works of course only if you have the clock signals connected as written in the manual. If not, the boards won't work after you switch the reference clock.
Thank you so much for the fast reply! I'll give it a try.